MACHEREY-NAGEL NANOCOLOR T-SET for heating block
NANOCOLOR T Set for electronic temperature control and calibration of the heating blocks NANOCOLOR VARIO 4 / VARIO C2 / VARIO HC / VARIO 3 / VARIO compact.
Includes CD with software, certificate and manual
NANOCOLOR T-Set - Electronic temperature sensor for
- inspection equipment monitoring
- analytical quality control according to GLP
inclusive NANOCOLOR T-Set software for
Electronic temperature sensor for temperature control and calibration
- recording of heating curves
- transfer of test and calibration data to PC
- creation of certificates and protocols
- programming of heating blocks
For consistent analytical quality control in water analysis the accuracy of the temperature of heating blocks has to be checked. With the NANOCOLOR T-Set the programmed temperatures of the heating blocks from MACHEREY-NAGEL can easily be tested and calibrated in case of deviation.
Temperature control: All temperatures programmed in the heating block are measured by the NANOCOLOR T-Set, displayed and stored in the heating block.
Calibration: All temperatures programmed in the heating block are measured by the NANOCOLOR T-Set, displayed, calibrated and stored in the heating block.
Functions T-Set software
With the NANOCOLOR T-Set software heating curves can be recorded to check the temperature stability of the heating blocks and temperatures and heating times can easily be programmed in the heating blocks via PC. Furthermore, test and calibration data measured with the NANOCOLOR T-Set can be transferred to PC. Certificates and protocols are issued automatically by the NANOCOLOR T-Set software just by pressing one button.
||PT 1000 (95 x ø 4 mm)
+ 0.1 °C
||via LED-Displays of the heating block and NANOCOLOR T-Set software
||via foil-covered symbol keys of the heating block and NANOCOLOR T-Set software
||0 - 200 °C
+ 0.2 °C
+ 0.1 °C
||bidirectional serial RS 232
||0 - 50 °C, up to 90 % relative humidity
||via RS 232
||max. 20 mW
||112 cm (length) including cable
||approx. 60 g
||calibrated with standard calibration thermometer
Declaration of conformity:
This instrument conforms to the following directives:
-89/336/EC of 03.05.1989 (including Amendment Directive 92/31/EC) - EMC Directive